VIRUS units continue to multiply! Shown below is a representation of 52 units on the sky:
The grey squares with stars on them are looking at the sky (with more than 23,000 fiberoptic cables taking simultaneous spectra!), and the white squares are empty slots for future units.
The VIRUS units (grey squares) do not actually touch each other, but are shown doubled in size to fill in the gaps between them, graphically.
Much like it is for the blogger, it is a real milestone for VIRUS to hit 50. In the case of VIRUS it is 50 units. This was achieved over this past weekend after spending several months with just 47 units.
Reconstruction from 50 IFU/VIRUS units on the sky. Each small square is 50 arcseconds on a side. There is actually a gap of 50 arcseconds between each IFU/VIRUS unit but it has been compacted in this view to avoid to much white space.
This means with every 6 minute exposure we take on the sky we get over 22,000 spectra!
On Monday we installed 2 more units bringing us to 52!!! Well on our way toward HETDEX science!
This week at the HET we have taken the top end of the telescope so that we can rework the layout of the IFU fibers and add 11 new fibers. That will leave just 17 IFU fibers to be installed. Just to be clear this is not spectrographs being installed but just the IFU fibers that feed future spectrographs. At the moment we have 41 spectrographs units installed and stable. This means we are not more than half way towards our total goal of ~78 units.
It has been a while since I posted and I think one of our biggest pieces of new is that we are now regularly observing with HPF. Not only is this instrument being used to observe host stars for planets but there are astronomers who are using it to observe planetary nebulae looking for new elements being born in the deaths of stars just slightly heavier than our sun. This new instrument is easy to use, reliable and a great addition to the HET.
The other big news is that we now have 40 double barreled spectrographs installed within VIRUS. That means that we are now more than half way to have VIRUS fully populated.
This week we had a number of break through moments. Perhaps the most prominent is that after weeks of rewiring and adding new shielding we may finally be at a point where the VIRUS array is fairly stable. We have 36 VIRUS units installed (each unit is 2 spectrographs) so a total of 72 spectrographs up for the last 5 days.
Another big breakthrough is in observing automation. We have a piece of software called OCD (Observatory Conditions Decision tool) which monitors the metrology (seeing, transmission and sky brightness) from the telescope and decides if HETDEX observations can be made. If allowed it will also take over the control of the TCS and VIRUS instrument and start observing HETDEX targets with only brief pauses to get confirmation from the telescope operator that we are on the right setup star. Using this new tool we were able to observe a full night of HETDEX targets with a minimum of overhead (even a few targets with just 3 minute setup times). This is faster than the fastest resident astronomer and telescope operator pair working together. Congratulations to the HETDEX software team who traveled out from Germany to help us get it all working efficiently.
In the last week we have had a few updates for two of our instruments. For VIRUS we are up to 29 working spectrographs. We actually have several more units but some of the oldest and slightly mis-behaving units have been sent back to Austin for realignment and recommissioning. For HPF we are thrilled to announce that a NIST laser comb has been installed in the calibration room in the HET basement. This allows us to send a picket fence of spectral features through a separate fiber next to the science fibers. During the data reduction and analysis of the HPF spectra they can look at the position of these pickets and determine how the instrument might be subtly moving and correct for it. So far it seems like it is working great.
A brief advertisement before we get into this week’s news: One of our Resident Astronomers is retiring. If you are a PhD astronomer and interested in working on one of the largest optical telescopes in the world doing cutting edge research please consider applying: https://utdirect.utexas.edu/apps/hr/jobs/nlogon/171020014209
This week the big changes to the telescope have been the removal of the Prime Focus Instrument Platform (PFIP) work platform and the arrival of 3 more VIRUS units. The PFIP work platform, as the name suggests, allowed us to drop people off to work on the PFIP. We will still be able to work on the PFIP but now we have to do it from the JLG worker-lift. The work platform covered up a number of mirrors (about 10% of the light collecting area) as can be seen in the images below taken with our pupil viewing camera.
Two images of the HET pupil taken with our pupil viewing camera before (left) and after (right) the PFIP work platform was removed.
With the delivery of 3 more VIRUS units we now have 28 working VIRUS units or 56 spectrographs or more than 12,000 fibers on the sky!
This week we are pleased to announce that a new VIRUS unit was installed in side two of the VIRUS enclosure. This brings us to 22 VIRUS units or 44 spectrographs. We also took a little time in the last engineering run to add on some valves to the vacuum fittings which will allow us to cold pump on the VIRUS units which takes far less time to do than to warm up and then repump which was our older methodology. Keeping 22 VIRUS units going is starting to be a little easier but still takes a lot of management.
In addition to the work on VIRUS, we have also installed in the coherent fiber bundles for the HPF. These coherent fiber bundles will be used to setup stars on HPF science fibers. HPF will arrive in the coming weeks and we are very excited to get our first high resolution instrument on sky in the coming months.
We are in another 3 week science period but our big news this week is that we have been able to bring a few more VIRUS Units on-line which brings our total number of active units up to 21 units. Recall that a VIRUS unit is made up of 2 spectrographs so we now have 42 spectrographs on-line. These newest units are located in VIRUS enclosure 2 which means that we are now making use of both of the “saddle bags” which were installed at the HET as part of the Wide Field Upgrade. This makes the VIRUS closer to being ready for the main part of the HETDEX survey. As we add more Spectrograph units the system continues to get more complex with multiplexers and timing systems.
In some recent very clear nights we were able to observe 10 HETDEX shots (what we call a pointing for the survey) in a single night. This is a new record and our setup times are now typically below 5 minutes when moving from one shot to another. Further small milestones….
The next few days are going to be very busy at the HET. The HET team along with help from Austin will be installing IFU’s for the second virus enclosure in preparation for more virus units to be installed. This involves stringing fiber cables from the enclosure up to the top of the tracker to be attached to the IHMP (Input Head Mounting Plate). The team will also be investigating and trying to remedy problems with the IHMP dither mechanism. During this time there will be no science or engineering related night work due to immobility of tracker, structure or dome for the IFU deployment.